"Red Rain" on ABI377 Gel File
tomlinson at pplros.demon.co.uk
Sun Jun 30 14:51:33 EST 1996
Yes, I've seen it. Red rain (according to ABI) is caused by dirty
The gel lifts slightly at the end of a long run, and is no longer
cooled, so the laser
burns holes in it (true, apparently!).
The solution is to periodically (say, every 10 gels) wash the plates in
NaOH, then in 0.1M HCl, which lifts off any crap that has adsorbed to the
from tap water etc. Also, something which worked for me is to lower the
temp to 48 degC. I've seen no rain at all since.
More information about the Autoseq