Temp fluctuations on ABI377
smarsh3807 at aol.com
Sat Dec 8 01:09:29 EST 2001
>Has anyone ever seen temperature fluctuations on an ABI377?
It could just be a loose connection to the the Temp sensors on the back of the
heat plate. Had it been serviced recently? If not try doing the total reset of
the board by pushing the reset button on the back twice and reloading the
firmware. If that does not fix it, it could be that the board is going out.
Director DNA Sequencing Core Facility at CalTech
More information about the Autoseq