[Automated-sequencing] RE: [SPAM] - Autoseq Digest, Vol 6,
Issue 3 - Email found in subject
ciaran.Fulton at arradx-almac.com
Thu Nov 17 11:24:20 EST 2005
Hi Philip...ironically we were just discussing the same error message that occurred on our 48-cap (3730) machine a week ago. This happened after the first half of the plate had been processed. The water tray had moved under the array and the machine just hung after generating this error message which was rather bizarre given that it was processing the same plate. No instrument warning light was observed and the error hasn't occurred subsequently. I'm sorry I can't shed anymore light on the matter Philip, nor have ABI technical support (I'm still waiting for them to return my call).
From: autoseq-bounces at oat.bio.indiana.edu
[mailto:autoseq-bounces at oat.bio.indiana.edu]On Behalf Of
autoseq-request at oat.bio.indiana.edu
Sent: 16 November 2005 17:00
To: autoseq at magpie.bio.indiana.edu
Subject: [SPAM] - Autoseq Digest, Vol 6, Issue 3 - Email found in
Send Autoseq mailing list submissions to
autoseq at net.bio.net
To subscribe or unsubscribe via the World Wide Web, visit
or, via email, send a message with subject or body 'help' to
autoseq-request at net.bio.net
You can reach the person managing the list at
autoseq-owner at net.bio.net
When replying, please edit your Subject line so it is more specific
than "Re: Contents of Autoseq digest..."
1. DNA SEQ: 3730XL Tray on deck error (Phillip San Miguel)
Date: Tue, 15 Nov 2005 13:45:18 -0500
From: Phillip San Miguel <pmiguel at purdue.edu>
Subject: [Automated-sequencing] DNA SEQ: 3730XL Tray on deck error
To: bionet-genome-autosequencing at moderators.isc.org
Message-ID: <437A2CBE.4090708 at purdue.edu>
Content-Type: text/plain; charset=ISO-8859-1; format=flowed
We keep running into a maddening error on our 3730XL:
Tray on deck does not match tray type in run setup
Maddening because this error usually happens in the middle of a batch of
runs from a 384-well plate. That is, the first or first few quads get
loaded and run fine. Then the batch errors out.
Usually it requires a service call to get the machine running again. But
I'm not entirely sure that the service call is what "fixes" the problem.
It might be turning the machine off for a couple of days. (An hour isn't
long enough. I just tried it.)
Replacing the plate sensor is among the methods that gets us running
again--for a time. But looking through old 3730.log files I can see 4
times that this has shut us down for a couple of days since the middle
of the summer.
Last time it coincided with our laser going out. The laser was replaced
and we ran for four days. Only to see this error pop up again.
Autoseq mailing list
Autoseq at net.bio.net
End of Autoseq Digest, Vol 6, Issue 3
This e-mail is from ArraDx Ltd
The e-mail and any files transmitted with it are confidential and privileged and intended solely for the use of the individual or entity to whom they are addressed.
Any unauthorised direct or indirect dissemination, distribution or copying of this message and any attachments is strictly prohibited.
If you have received the e-mail in error please notify helpdesk at arradx.com or telephone +44 28 38 363841 and delete the e-mail from your system.
E-mail and other communications sent to this company may be reviewed or read by persons other than the intended recipient.
Viruses : although we have taken steps to ensure that this e-mail and any attachments are free from any virus, you should, in keeping with good practice, ensure that they are actually virus free.
ArraDx Ltd. Registration Number NI 43067
Registered Address : Almac House, 20 Seagoe Industrial Estate, Craigavon, BT63 5QD
More information about the Autoseq