Mysterious signal intensity variation is plaguing us. Even capillaries
show good signal strength, odd capillaries show poor signal strength. I
also see this with ABI sequence standards, so it is not a problem with
Here is a screen capture of a standard run:
Not sure if it is relevant, but here is a screen capture of our spatial
I hand-loaded the sequence standards myself into the A1 quad of an ABI
384 well plate and heat sealed it myself. (By which I mean there are no
One of the service engineers attempting to troubleshoot this over the
phone thinks it is a result of the array electrode being pressed against
the bottom of the sample plate during electrokinetic injection. That,
plus a slight difference in the length of the caps that maps even/odd.
He had us try to back off a z-axis calibration in the 3730calib.ini file
by 20 units. That resulted in the autosampler being unable to get the
sample plate. So we tried 10 units. The autosampler worked okay then,
but the "picket fence" phenomenon remained.
Although this is a new array, in looking back at old runs, I think I see
the same phenomenon in previous arrays--just not as pronounced.