Fulton, Ciaran wrote:
> Has anyone any recommendations for sealing 384-plates not using a
> sealer. We currently use adhesive foil seals from ABgene which seem Ok
> but our ABI engineer thought they might interfer with the array.
>> Any comments?
>You might get adhesive residue on the piercing nails. That's the only
reason I'm aware of that ABI switched from recommending foil seals
(3700) to heat seals (3730).
ABgene has a relatively inexpensive heat sealer (I don't mean the
$20,000 monster recomended by ABI). It works fine. It is even possible
to remove the heat seal film and reseal (if you don't over seal the