[Bioforum] working temperatures for Porapak Q 80-100 C-5752 to work
with liquid ethanol 99.5% enquiry
(by kloyake from hotmail.com)
Fri May 29 17:08:52 EST 2009
I am wondering if I can use liquid ehanol 99.5% as a standard to work with GC (Shimadzu GC-14B) and Porapak Q 80-100 C-5752.
If the answer is yes, what is the column temperature, injection temperature and detector temperature I should work with?
Now I was told to dilute the ethanol with DI water to the concentration that suit my sample (ppm). The GC set to column temperature 140 C, injection temperature 160 C and detector temperature 180 C. After some trials I found the results of the peak areas were not consistent with the concentrations. Please advice anything that might suit my work. I have to finish this work very soon.
Insert movie times and more without leaving Hotmail®.
-------------- next part --------------
An HTML attachment was scrubbed...
More information about the Bioforum