Microinjection experts; help me out please!
Ary Correa Jr.
ac24 at cornell.edu
Sat Feb 25 10:54:41 EST 1995
In article <3il6de$6ud at news.iastate.edu>, letrud at iastate.edu (Louis-Eric
> umdoble at mail.cc.umanitoba.ca (Brad Doble) wrote:
> >I am trying to inject cultured neonatal rat cardiac myocytes, and have a
> >problem with the microinjection pipette surging forward when I press the
> >inject button (often leading to ruined pipettes!). This happens even when
> >the injection pressure (as displayed on the LED readout of the injector)
> >is only 1 psi. Does anyone know what might be going wrong? I am using a
> >fluorescent dye to monitor the delivery of the injectant and thus, know
> >that the 1 micrometer diameter tips of the pipettes are not clogged.
> >Any feedback would be greatly appreciated.
> >Thanks in advance,
> >Brad Doble umdoble at mail.cc.umanitoba.ca
> > or: bradd at sbrc.umanitoba.ca
> >Department of Physiology, University of Manitoba
> One possibility is that the diameter of the opening of your capillary holder is too large for the O.D. of your pipettes.
> Another possibility is the clamping of your pipette in the holder: is there a suitable O-ring there?
> Good luck.
> Laboratory of cellular signaling
> Iowa State University
My guess is that you are compressing the O-ring in the pipette holder while
injecting. Try a wider capillar tubing or replace the O-ring for a
thougher one. If you are controling the pipette movement manually, you can
try to back the pipette wilhe injecting. Takes some skill but can be done.
More information about the Cellbiol