derek_davies at postmaster.co.uk
Mon Jul 6 01:48:56 EST 1998
In article <6nhvcd$ajb$1 at nnrp1.dejanews.com>, darren at indy.bio.uts.edu.au wrote:
> I have a colleague who wants to analyse algal cells using our FACS machine.
> Basically, he wants to excite the algal cells at one wavelength and detect at
> another, which is as I understand it how FACS works- it excites at 488nm and
> detects up to 3 (ranges of) wavelengths . However, he wants to vary the
> excitation wavelength to eg 640nm and then detect flourescence at say
> 670-700nm. Is this plausible/possible? I don't think it is on our machine (We
> have a BD single laser FACScan) but maybe on a different model?
> Thanks, Darren Jones
You are right, it isn't possible on a single laser FACScan. To excite at
you will need a red laser. The dual laser FACS Calibur comes with a 635nm red
diode laser which would do the job. The alternative is to fine a sorter
equipped with a Helium-Neon laser (again emitting at 635nm). The collection
filters of both machines can be set to the specific wavelength required.
Derek Davies, FACS Laboratory, ICRF, London.
More information about the Cellbiol