[Cell-biology] old centrifuge without a speed indicator
(by R.Jayakumar from roswellpark.org)
Thu Apr 19 13:17:09 EST 2007
That is what it exactly is - a clinical centrifuge.. Not an analytical
one. This one obviously is only used for spinning down cells. Run a
few of your cell samples at different speeds and check your yield and
pellet consistency after 5 min. The pellet should easily suspend upon
tapping. If it is too firm, the spped is too much. Use that speed for
your cell lines.
From: cellbiol-bounces from oat.bio.indiana.edu
[mailto:cellbiol-bounces from oat.bio.indiana.edu] On Behalf Of Porpyrin
Sent: Thursday, April 19, 2007 11:39 AM
To: cellbiol from magpie.bio.indiana.edu
Subject: [Cell-biology] old centrifuge without a speed indicator
Recently, I got an old centrifuge for my tissue culture. A problem is
there is no clear indicator for a speed. There is just numbers from 1 to
The model is:
DAMON IEC DIV ~ CLINICAL CENTRIFUGE
Does any body know how can I know the speed from the numbers? I need to
set the speed at 1000-2000 rpm.
Cellbiol mailing list
Cellbiol from net.bio.net
This email message may contain legally privileged and/or confidential information. If you are not the intended recipient(s), or the employee or agent responsible for the delivery of this message to the intended recipient(s), you are hereby notified that any disclosure, copying, distribution, or use of this email message is prohibited. If you have received this message in error, please notify the sender immediately by e-mail and delete this email message from your computer. Thank you.
More information about the Cellbiol