jsp at bio.aucNOSPAM.dk
Mon May 12 15:37:47 EST 2003
I have a SpectraMAX Gemini XS platereader working perfectly here in Aalborg.
The only problem I have experienced is that some times the cut-off filter
wheel is stuck which causes a stop in kinetic readings followed by an error
#324 message. Disabling the cut-off filter eliminates this error. We were
not pleased to find that replacement of the flash-lamp cost us 10.000 Dkr
(1300?), but the lamp did indeed last for many reads during long kinetic
"Kasper Engkilde" <kde at mail.tdcadsl.dk> wrote in message
news:BAD35A7B.73BD%kde at mail.tdcadsl.dk...
> Hi all
> I'd like to hear if anyone have had experiences with malfunctioning
> fluorescence plate readers from Molecular Devices (Gemini series). We
> one a year ago and have had a lot of trouble with them. We need one more
> would like to hear if we were simply unlucky (as MD claim) or their
> instruments are generally unstable.
> Best regards,
More information about the Fluorpro