William B. Melchior, Jr.
wmelchior at fdant.nctr.fda.gov
Wed Dec 14 15:06:28 EST 1994
In article <1994Dec12.140127 at opal.tufts.edu>, kmorris1 at opal.tufts.edu says:
>The red liquid in our thermometers seems to have "split"... ie, there are
>several small "pieces" no longer attached to the larger reservoir. The
>thermometer, as a result, no longer reads accurately. Is there any way to
>"reattach" the whole column of liquid together?
Several people have suggest deep chillling, which is the preferred method.
At least with mercury thermometers there is an alternative, which I think of
as the method of LAST RESORT, since it may break the thermometer:
Get a narrow tube a yard or so long and rest it vertically on a firm but not hard
surface -- I use a notepad at least 1/4 inch thick. Drop (!) the thermometer, bulb
end first, down the tube so it jolts against the pad of paper. This may have to
repeated several times.
I have never actually broken a thermometer doing this, but I may have been lucky.
If pressed, I may have to deny ever having written this message. (The only time
this method has failed for me is when the interior capillary has become dirty
More information about the Methods