Protean II glass breaking
Xiao-Ru.Wang at genfys.slu.se
Fri Nov 25 08:22:15 EST 1994
In article <Czs71y.F1o at cc.umontreal.ca>, cohenea at ERE.UMontreal.CA (Cohen Eric) says:
>This is our problem:
>We do an awful lot of SDS-PAGE work and we use the Biorad protein
>electrophoresis Protean II xi system. Sometimes, the glass plates
>or even the sandwich clamps that hold the plates break in the middle of
>the run. After trying out the cooling systems, we still have this
>problem from time to time.
>Eric A. Cohen, Ph.D.
>Montreal University School of Medicine
We have the same problem with Biorad´s Protean II xi system. Glass plats
break very often during sandwich set-up or during the run. We cope with the
problem by visiting a glass work-shop to cut 50 spare plats for lab storage.
Dept. of Forest Genetics & Plant Physiology
Swedish Univ. of Agricultural Sci.
More information about the Methods