ABI sequencer, a month of unexplicably bad results.
vblasch at gwdg.de
Fri Dec 22 04:05:37 EST 1995
We had a problem with the 373 too, due to an electric shortcut in the
system which did a reset on the photomultiplier settings. As a result,
the signals were too weak to interpret and resulted in almost unusable
results. Everything was fine again after reprogramming. Since your
mistake disappeared by itself and an ABI sequencing reaction worked fine,
it would argue against this. However, you might want to consider this.
More information about the Methods