drm21 at mole.bio.cam.ac.uk
Tue Oct 21 16:02:01 EST 1997
In article <199710211522.JAA22722 at nestor.NMSU.Edu>, hroychow at NMSU.EDU
(Hiranya Roychowdhury) wrote:
> We do not have it but would like to try and build one. A while back
>there were a few postings on where to find the schematics or how to build
>one from stuff off the local Radio Shack. Can anybody in the net point to
>that direction? My UG project student would like to try making an e-porator.
Take a look at Dower et al, Nucleic Acids Res Vol 16, pp6127-6144 "High
efficiency transformation of Ecoli by high voltage electroporation", 1988.
This includes a simple circuit and some discussion of the parameters.
I used to have another reference (from before the end of 1991, maybe
Biotechniques?) that described a way of making a LOW voltage
electroporator. Essentially, what is impportant is the voltage _gradient_
not the absolute voltage. Thus, if you drop the electrode separation
10-fold, you can drop the supply-voltage 10-fold. IIRC (which I may not)
the authors achieved a very small electrode gap by sticking insulating tape
onto a sheet of stainless-steel (?) and could then use a regular
electrophoresis power supply as the source. There was some very simple
capacitor/resistor circuit with one or two push-buttons to allow charging
of the capacitor from the power-supply and (exponential) discharge through
I have a vague recollection that I might have posted the reference here
_several_ years ago, so maybe a search with my name would help?
PS Make sure he doesn't kill himself!
D.R.Micklem, Time flies like an arrow...
Wellcome/CRC Institute, Fruit flies like a banana.
Cambridge CB2 1QR, UK Email:drm21 at mole.bio.cam.ac.uk
Unsolicited mail will incur a US$100 processing charge.
More information about the Methods