jorge2 at earthlink.net
Thu Jun 25 22:01:21 EST 1998
il128 at my-dejanews.com wrote:
> I have an autoclave validation problem and was hoping to get some input from
> my fellow netwise microbiologists. The cycle is a prevac with a 60 minute
> dwell time at 121.1C at 15psi. The load is made up of 50 bags (Tyvek
> Breather bag) of stoppers each bag is about half full. Some of the stoppers
> (the indicator stoppers) are sticking together and when they do B.
> stearothermophilus spores are surviving. The stoppers are washed and
> siliconized before spiking, bagging and autoclaving. Any suggestions on how
> to prevent the stoppers from sticking together? Paul
> -----== Posted via Deja News, The Leader in Internet Discussion ==-----
> http://www.dejanews.com/ Now offering spam-free web-based newsreading
Why 60 minutes?
More information about the Microbio