About Pipette Puller
cje2 at york.ac.uk
Tue Aug 22 05:30:08 EST 1995
Try using some platinum wire instead, and make a coil former out of a helical
groove cut on a cylinder with the ends cut down to the diameter of the glass
you use. then you can slip the former in and check the coil size
In article <199508192354.QAA09527 at net.bio.net>,
LIUT%SNYBKSAC.BITNET at UBVM.CC.BUFFALO.EDU (Taosheng Liu) wrote:
> Does anyone have experience with David Kopf vertical pipette puller?
>I use a model 720 with a nichrome heating coil, but the coil always changes
>shape after several heatings so it's impossible to get the right electrdes.
>doing two-stage pulling to do whole-cell patch clamp.
> Any suggestions will be appreciated (it's unlikely for us to buy a new
>puller). Please reply or e-mail me(liut at sacc.hscbklyn.edu)
More information about the Neur-sci