Reply to: RE>re: EEG LIST
Dr. Knowles is exactly right - most animal grade front ends will fail
the 10 microamp leakage test. An alternative is to use a clinical
grade isolation transformer.
Now don't try this at home, kids!
Ira Rampil, MSEE, MD
Medical Director, Neurophysiologic Monitoring Service